In the resulting window you can:
- Run automatic analysis by clicking the icon;
- Apply 2-marker (2-point LSA attenuation) or 5-marker (5-point attenuation) mode by clicking the corresponding buttons in the top right corner;
- Edit events in the 2-marker or 5-marker mode by clicking the corresponding buttons:
Please note that the buttons become active depending on what mode is selected.
After that you can apply your reference trace to the other traces from the same cable.